Combining clinical excellence and compassion
What can you tell us about your background?
I was born and raised in Baltimore, Maryland. I attended college, dental school, and my postgraduate endodontic residency at the University...
What can you tell us about your background?I grew up in southern Orange County and earned a bachelor’s degree in biology at San Diego State University in 1983. There I met my future wife, Kim, at the...
Focus on family, patients, friends, growth, and community
What can you tell us about your background?
For as long as I can remember, I wanted to make people smile. I always loved getting the class laughing in grade school. Looking back, I am sure...
Dr. Robert Slosberg facilitates accurate mapping and obturation of the resportive defect with CBCT imaging
AbstractA patient presented with advanced internal root resorption of tooth No. 9. The prominent location of this tooth...
Drs. Brian Shaughnessy, Margaret Jones, Ricardo Caicedo, Joseph Morelli, Stephen Clark, and Ms. Jennifer Osborne review the occurrence of teeth presenting with condensing osteitis and their associated pulpal diagnosis over a 2-year period.
Dr. Andrei Zoryan dispels some of the common myths surrounding carrier-based obturation
Carrier-based gutta percha
Carrier-based obturation (such as Thermafil®, GT® obturator, ProTaper® obturator [Dentsply Tulsa Dental Specialties]) is one...
In part 2 of his series, Dr. Ace Goerig suggests ways to reduce stress in the practice
Almost all endodontists could be completely out of debt and on the way to financial freedom within 5 to 7 years if they only knew the secret. But the secret is...
Dr. Robert Fleisher ruminates on how to prepare for retirement
There are so many articles about everything that you become pretty much overwhelmed and can never expect to read them all. So you pick and choose. You like to learn about the latest and...
Dr. Roger Levin presents the 10 top ways to help create a perfect dental team
With the changes brought on by the economy, top companies are bringing in the best resources they can find to evaluate where their organizations stand. They want to know...
Educational aims and objectivesThe purpose of this article is to explain how to remove a fractured instrument with a new type of extractor.Expected outcomesCorrectly answering the questions on page 35, worth 2 hours of CE, will demonstrate you can describe how to remove a fractured instrument with a new type of extractor. Click here to take the quiz to receive CE credits.
IntroductionThe fracture of a root canal instrument during endodontic treatment is a surgical accident that every practitioner is likely to have encountered at some point during their careers. The risk of instrument fracture has, however, increased in recent years in line with the constantly growing use of rotary instruments for root canal preparation. A recent study carried out on the practical experience of seven endodontists over a period of 13 years has revealed a risk of instrument fracture of 3.3% (Spili, Parashos, Messer, 2005). Of those, 78.1% of the fractured instruments turned out to be rotary instruments made of nickel titanium. A number of factors were associated with the fracture of nickel-titanium (NiTi) instruments, including:• The experience of the operator (Yared, Bou Dagher, Kulkarni, 2003)• Rotational speed • The degree of curvature of the root canal• Instrument design • Application technique • Manufacturing procedure • Non-patency of the root canalThere are two different situations that typically result in the fracture of rotary NiTi instruments: 1. Torsional fracture 2. Fracture due to cyclic fatigue Both fractures are associated with the treatment of extremely curved canals. NiTi instruments fracture most frequently because the instrument tip is subjected to excessive torque when the helical blades penetrate into the radicular dentin. The fragment, which usually has a length of 3-4 mm, remains jammed in the radicular dentin. The other characteristic criterion of these fractures is associated with the instrument design. Most of the rotary NiTi instruments have a taper of 4% to 12%. Due to this increased taper, which clearly exceeds that defined by the ISO standard (2%), the coronal part of the fragment remains blocked in the canal whereas its tip remains free. This particular feature–typically found in NiTi instruments–complicates the procedure of removing the fragment, which normally entails passing an endodontic hand instrument between the fragment and the canal wall and guiding it along the fragment to regain patency of the canal (Suter, Lussi, Sequeira, 2005). In this case, a more invasive solution is required. This involves straightening the coronal curve and releasing the fragment at the expense of the dentin walls. It has been shown that the attempt of removing these fragments leads to the removal of large quantities of dentin, thus resulting in a significant reduction of the mechanical resistance of the root (Spili, Parashos, Messer, 2005). A multitude of tools designed to remove fragments of fractured instruments have been developed, with the Masserann set by Micro-Mega being the most well-known. This efficient kit contains an extractor that seizes the fragment before removing it. The main disadvantage, however, is the size of the extractor (external diameter 1.20 mm), which requires the canal to be enlarged up to the size of a Gates Glidden drill size 5 before the extractor can be inserted. Although this extractor is very useful for removing fragments lodged in the coronal part of large teeth, it is unsuitable when it comes to removing a fragment located in the middle or apical third of the root canal or if the fragment is located in the thin root of a molar. The introduction of the operating microscope resulted in the development of ultrasonic tips specially designed for this specific use: • ET 25 (Satelec)• ProUltra® endo 6, 7, 8 (Dentsply)• RT3 (EMS) The principal task of these ultrasonic tips is to extract the fragment through vibration (Ruddle, 2004). Although effective, this technique is very time-consuming and requires an operating microscope. What’s more, it demands great skill on the part of the practitioner, and it necessitates the removal of a considerable amount of radicular dentin (Alomairy, 2009). A new set has recently been introduced (Endo Rescue Kit 4601, Komet) (Figure 1), which–though following the same basic approach as trepan burs, i.e. the exposure of the instrument fragment at the expense of the dentin walls–includes a number of special, advantageous features. The first feature is the presence of a center drill (Figure 2) with the same diameter as the trepan bur. The task of this center drill is to prepare the site for the work of the trepan bur. The other special feature is the miniaturization of the trepan bur. The external diameter of the trepan bur is merely 0.9 mm, which corresponds to that of a Gates Glidden drill size 3 (Figure 3).
Clinical sequenceThe clinical procedure consists of three consecutive steps:First step:The aim is to achieve direct access to the fragment while removing as little dentin as possible. Instrument fractures in the coronal or middle third of the root canal are most frequently caused by incorrect preparation of the access cavity or the presence of a curve in the coronal third of the canal. The first step is, therefore, dedicated to rectifying the canal access and relocating the coronal entrance. This requires straightening of the canal wall opposite the curve. The instruments recommended in the Endo Rescue set for this step are those that are usually suggested for this task: • A cylindro-conical bur with non-cutting tip for straightening the walls of the access cavity• A short Gates Glidden size 4 for relocation of the canal entrance, which is moved along the wall opposite the coronal curve with vertical back-and-forth movements • A Gates Glidden size 3 to prepare access to the fragment (clinical case Figures 4 and 5).
The aim of this step is to expose the coronal part of the fragment and re-center the access passage to the fractured instrument. A center drill (RKP.204.090) has been specially designed for this purpose. This center drill has the same external diameter as the Gates Glidden size 3, but the active part at its end has a tapered, concave shape. The outer blades drill into the dentin surrounding the fragment, and the concave, tapered area that encounters the coronal part of the fragment allows centering of the preparation. Thanks to the small diameter of the instrument, this task can be carried out removing just a minimum of dentin, while working in the center of the canal.
LimitationsThe Endo Rescue Kit reaches its limits when the fragment is particularly long or located beyond the curve of the canal, in which case the blades of the trepan bur expose the coronal part of the fragment until the stop position is reached. The action of the trepan bur is interrupted before the fragment can be released from the dentin in the root canal. In this situation, the trepan bur can be used in combination with a suitable aid, such as an ultrasonic tip or a cannula filled with composite. As detailed above, the trepan exposes the coronal part of the fragment, but this is followed by a cannula of the same outer diameter as the trepan bur (i.e., size 20) filled with autopolymerizing composite, which is placed onto the already exposed part of the fragment. Once the composite has hardened, the cannula is pulled out in counterclockwise direction. The fragment that has now adhered to the hardened composite is pulled out along with the cannula (Figures 10-15).
DiscussionThere are numerous tools and methods for the removal of instrument fragments fractured in the root canal, however, most of them produce unreliable results and cause significant damage to the remaining radicular structure. Ideally, the radicular dentin should be preserved as far as possible, and the extent of the root canal preparation after removal of the fragment should not exceed that of a conventional preparation. Although the Endo Rescue system requires the removal of an additional amount of dentin, the reduced diameter of the instruments allows the fragment to be accessed with minimum damage to the root structure.
In line with all previously described methods, the decisive factor for a successful intervention is the preparation of the access to the fragment. Due to the fact that instruments fracture most frequently in the canal curve, it is vital to straighten the coronal curve in order to achieve direct access and an unobstructed view of the fragment.A special situation that is particularly associated with the fracture of rotary NiTi instruments should be noted. Due to the fact that the helical blades of the instrument penetrate the radicular dentin in a rotary motion, thus generating a torque that exceeds the resistance of the instrument, most of the fragments are of small diameter, reduced length, and above all, completely surrounded by dentin at the site of the fracture. If ordinary ultrasonic tips are to be used for the removal of the fragment, this should take place without touching the fragment to avoid further fracture. This is a very tricky and delicate procedure. However, by centering the access passage to the fragment (center drill), exposing the fragment by removing the surrounding dentin (trepan bur) and then pulling it out in an counterclockwise direction, this tool is ideal for removing fragments of NiTi instruments–which currently constitute the majority of instrument fractures. At the moment, no other tool designed for dealing with this specific situation is available on the market. The limitations of this system are associated with the root canal anatomy and the feasibility of achieving the required straight access to the fragment. An instrument fragment located in the apical third beyond an extreme curve cannot be removed with the Endo Rescue Kit alone. In such cases, it is recommended to use the trepan bur in combination with a cannula of identical diameter filled with composite.
ConclusionThe Endo Rescue Kit described in this report aims to provide a standardized procedure for the removal of fractured instruments. Compared to other instruments available for this purpose, it not only significantly reduces the amount of dentin that has to be removed in order to pull out the fragment, but also the amount of time needed for this procedure. AcknowledgementsThis article was first published in L’information dentaire, issue no. 11 (March 2010), and has been republished with permission.
ReferencesAlomairy KH (2009) Evaluating two techniques on removal of fractured rotary nickel titanium endodontic instruments from root canals: an in vitro study. J Endod 35(4):559-562.Ruddle CJ (2004) Non-surgical retreatment. J Endod 30:827-845.Souter NJ, Messer HH (2005) Complications associated with fractured file removal using an ultrasonic technique. J Endod 31(6):450-452.Spili P, Parashos P, Messer HH (2005) The impact of instrument fracture on outcome of endodontic treatment. J Endod 31(12):845-850.Suter B, Lussi A, Sequeira P (2005) Probability of removing fractured instruments from root canals. Int Endod J 38(2):112-123.Yared G, Bou Dagher F, Kulkarni K (2003) Influence of torque control motors and the operator’s proficiency on ProTaper failures. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 96(2):229-233.
Subscribing to Endodontic Practice US magazine has never been easier.
1 Year Subscription $99.003 Year Subscription $239.00
Research has shown that irrigants are more effective when they are electro-mechanically activated.
Research has shown that irrigants are more effective when they are electro-mechanically activated.
Acoustic streaming and cavitation have been proven to significantly enhance cleaning of difficult anatomy. Studies have shown that low frequency (Sonic) oscillation (160-190Hz) was not sufficient to create acoustic streaming or cavitation within the canal space.
Take advantage of many web and print advertising opportunities.
Subscribe online, or call our team (866) 579-9496.
Receive CE credits through our website.
Contact our website news team via our Online Customer Support Service.